Pink noise or sweep measurement signal?
Both are good for different purposes. Noise is more robust against narrowband disturbances, such as hum or speech. Sweep is better for detecting non-linear distortion. In an environment without much disturbance, then a sweep is preferable.
What measurement signal does Dirac recommend for a specific speaker?
For most cases, a sweep is preferable.
What frequency response target curve does Dirac recommend for optimization?
Please see the Target curve design guidelines in the User Manual.
Can DLCT import target curve files?
Yes, in the Filter Design tab it is possible to load a previously saved target curve.
How do I manage a multi-room measurement in DLCT for simultaneous multi-room processing in Dirac Audio Processor (DAP)?
Filters can be designed for multiple listening rooms by creating multiple projects in DLCT. However, multi-room simultaneous playback is not yet a supported feature in the Dirac Audio Processor. If you are using an external processing unit, then multi-room may be supported by that device. To load several filters for simultaneous use in a multi-room setting, then please consult the user manual of the corresponding processing unit.
How does phase correction work above the Schroeder frequency?
Above the Schroeder frequency (or generally, the higher frequencies that you consider) the job of mixed-phase correction becomes trickier. First, you need to have a proper amount of measurements spread out over the listening region. This makes it possible to determine which non-minimum phase behavior that is consistent across measurement positions (and in between) and which behavior is inconsistent (that is, too different from one position to another to be possible to correct for). Second, you should only correct for the behavior which is consistent across the measurement positions. The higher in frequency you go, the more consistent the behavior has to be to be robustly invertible. In essence, the amount of phase correction that is applied is typically lessened with frequency (as is the level of detail in your magnitude response correction). This is typically the case, but all of this is determined from the actual measurements. The correction is always tailored to what is actually measured in your particular case. The trick is to do the most correction that you can robustly over all the measurement positions, but never more.
Why is there no validation feature?
The predicted response may look too good to be true, but it is important to interpret the curves in the right way; the prediction curves show the response in exactly the point each curve is based on. In theory the curve can be totally different just a small distance away from it, but in practice this is rarely the case. In order to avoid optimizing the filter for a really small area at the expense of the actual listening area, multiple measurements are used, taken in different points. Their average is the basis for the filter calculations. Together with the spatial robustness of the Dirac Live technology this will result in a filter that sounds good in a large area. However, for a measurement to produce the predicted curve the microphone need to be in the exactly same position, otherwise the result will just be close to the prediction. It is also important to compare a validation measurement to the curve that corresponds to that measurement position and not the average curve.
If you want to measure the result of the filter you can either use some application such as HolmImpulse, or you can select the Dirac Audio Processor as the output device on the sound setup page. If this method is selected it is very important to note that the Dirac Live Calibration Tool will disable filtering in the Dirac Audio Processor before each measurement, and you will have to enable it manually real quick. The reason the Dirac Live Calibration Tool disables filtering is that most measurements will not be for validation but for room analysis, and in this case any filtering has to be disabled.
How are the displayed frequency curves smoothed?
The displayed individual (select “All (before)” or “All (after)” to view) frequency response curves are smoothed with a 1/8 octave smoothing filter. The displayed average (do NOT select “All (before)” or “All (after)” to view) curves are averages of the individual (unsmoothed) frequency responses, then smoothed with the 1/8 octave smoothing filter.
What formats for microphone calibration files do you support?
We support UMIK-1 calibration files, XTZ calibration files and .omm calibration files used by Dayton Audio. There is currently no clear standard on how calibration files look, when there is one we will implement support for that. If you got a microphone with another calibration file, please read the documentation on how to adapt it for use with our software.
Avoid any combination of microphone and sound cards?
In general, it is difficult to say how different hardware and their driver software will work together on a computer system. It usually depends on how well they comply with standards. Simultaneous use of two USB audio devices must be through different USB controllers to work properly. Therefore, make sure to have a different USB controller for each USB audio device that will be used at any time.